Department of Electrical Engineering and Technology. Is there any additional driver SW from Agilent that you need to install to get the device to be recognized correctly? The procedure for detecting your device depends on the bus being used. Insert another Read Output. Click the Agilent EX project to create a new copy. How often does your dual instrument VI update the data? The power supply should turn on and depending on the text in the other field, you can change the settings.
|Date Added:||6 June 2015|
|File Size:||50.16 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
With compact and modular form factors, NI’s PXI programmable power supplies and precision sources simplify the task of designing automated test systems by saving expensive rack space and reducing the need for multiple instrument form factors within a test system. Figure 6 VI Function Generator Block Diagram The design of the various stages, functions, and blocks will not be explained within this lab but you should turn on the Show Context Help found under Help in the main menu so that when your cursor passes over objects in the front panel text will appear giving some information about what the blocks do.
It is also available from: Delete the wires connecting Configure Current Limit.
Agilent E36xx IVI-COM Driver
Agilent EA – error The mark LabWindows is used under a license from Microsoft Corporation. Wentworth Institute of Technology. Then go to the Block Diagram for the Power Supply. So I got myself a serial-usb-adapter and downloaded the driver from the NI website: Windows is a registered trademark of Microsoft Corporation labiew the United States and other countries.
Take a few measurements by changing the power supply voltage in the front panel VI for the power supply you could also connect the function generator to the DMM and measure an AC waveform. Turn on the labviee supply and you will see the resources are now increased by one; if not you will have to click on Refresh to see the new instruction in the VISA resource name drop down menu.
Application Programming Interface API To help accelerate software development, all instrument drivers follow a consistent programming flow of: Message 4 of However, I get the very same error when I plug my power supply into the serial port of my computer and try to run the test VIs.
Enable tracking on the ‘Configure Tracking’ function You can verify that tracking is enabled by reading the output values of the paired channel. Click on this folder then open the Examples folder and select the highlighted file in Figure 10 which opens another window: This block diagram also demonstrates the complexity of the advanced designs used to control test instruments. Add one final Concatenate Strings function to append all the sub-strings into one formatted string.
Connect the purple instrument wire and yellow error wires through the second Read Output. Known Issues This driver does not have any known defects. In the tutorials it is always basically just plug and play so I didn’t expect that I needed to install 363a software. You can download the latest version from: For the lab report, include the front panel and the block diagram for the combined VI in your submission, your observations and any questions you have been asked to respond to. Message 5 of Enable tracking on the ‘Configure Tracking’ lsbview.
EA – Want display to show voltage/current i | Keysight Community
Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or select the Learning tab from the top right. The power supply should turn on and depending on the text in the other field, you can change the settings. This is useful for circuits that need balanced positive and negative inputs.
Message 6 of You can download the parser and its associated SDK from: Using the agillent draw the cursor around all the items in the design. The very first function you need to use is the initialize. When the front panel window opens, name and save the VI in a location you can access again.